Skip to Content
The Massey Genome Service requires customers who are using the ‘Genotyping Capillary Separation Service’, to please provide your pooled reactions in a volume of 1ul, in either 0.2ml individual PCR tubes or 0.2ml strip tubes. For the ‘Genotyping Plate Service’, please provide your pooled reactions in a volume of 1ul, in a 96 well plate, sealed with strip tubes, plastic seal or foil seal. The strip tubes provide the best seals.
If is recommended that when you first set up your experiment to carry out a concentration titration series on your pooled genotyping products to determine optimal loading. Please contact Richard Fong for further information regarding optimising loading of products.
For genotyping with the Massey Genome Service customers use 5’-end labelled primers. Customers can use the following 4 ABI Prism® dyes to label their primers with:
The Massey Genome Service uses GeneScan™ - 500 LIZ™ Size Standard, as the internal standard for fragment sizing. This size standard will accurately size fragments between 50-500bp, and provides 16 single-stranded labelled fragments of 35, 50, 75, 100, 139, 150, 160, 200, 250, 300, 340, 350, 400, 450, 490, and 500 bases. Each of the DNA fragments is labelled with a proprietary fluorophore, which results in a single peak when run under denaturing or native conditions.
NOTE: Customers do not have to provide the size standard. The Massey Genome Service will add the size standard to your samples before running. When designing your primers, please design them to produce fragments >75bp and <490bp. The Massey Genome Service uses ABI Prism® GeneMapper™ Software which looks for two size markers smaller and larger than the fragment being sized, in order to get an accurate size calling. The 35bp size standard fragment can sometimes be inaccurate because the unincorporated primer peaks obscure its detection by the software. Also the 250bp fragment peak in the GeneScan™ - 500 LIZ™ Size Standard, is not used in fragment sizing under denaturing conditions, due to the abnormal migration of double strands that did not completely separate under these conditions. The 250bp fragment peak frequently runs at 246bp, so it is not included in size calling. So there is a gap of 100bp between the 200bp and 300bp size standard peaks when sizing fragments in this size range.
The following recommendations are provided to help optimize primer selection:
Please download the attached ‘Genotyping Technical Bulletin’ below, which contains information on the following:
Please download the attached ‘Microsatellite Fragment Analysis Technical Bulletin’ below, which contains information on the following:
Page authorised by Web Content Manager
Last updated on Friday 24 August 2018